Attenuated Total Reflectance (ATR)$99 Base price Go to quote list
White Light Interferometry
White light interferometry (WLI) is a nondestructive, non-contact, optical surface topography measurement which uses coherence scanning interferometry to generate 2D and 3D models of surface height.
- Steep cut-offs and sharp topography angles impair WLI measurement accuracy around these features
- Interpretation of fringe changes can be difficult in certain types of samples
3D WLI measurement: the figure of Lincoln sitting in his monument on the surface of a US 1 cent coin (barely visible with the naked eye), visualized clearly with WLI.
3D WLI measurement of a circuit board feature.
- Solid phase
- Material: opaque, transparent, coated, uncoated, specular, or rough
- Maximum lateral dimension: 147 mm
- Maximum vertical height: 100 mm
- Larger sample width and depth possible with partial coverage / scan area
- Sample reflectivity: 0.05 – 100 %
Zygo ZeGage Plus
- Magnification Range: 10x to 50x
- Maximum Vertical Scan Range: 20 mm
- Surface Topography Reproducibility: ≤ 0.15 nm
- Optical Lateral Resolution: 0.52 μm
- Step Height Accuracy: ≤ 3%
WLI instruments use a white light source to illuminate the surface of the sample.
A beam-splitter divides the white light beam into two optical paths: one that reflects or scatters from the sample and one that reflects from a flat, known reference mirror. These two signal beams are then mixed together and the resulting image is projected onto a CCD image sensor.
The two beams, when mixed, form an interference pattern whose intensity can be related to the sample surface height: any difference in optical path length between the reference and sample beams changes the measured interference intensity at each scanned point, providing an indirect measure of the height variance in the sample.
The interference fringes are analyzed at each point to build up a 3D map of the sample surface.
No resources available at this time.
Optical microscopy is ubiquitous in diverse fields within academic research and commercial industries. It is an affordable, rapid...
AFM measures surface topography and certain material properties with sub-nm vertical resolution and atomic-level force sensitivity.
Chromatic dispersion profilometry is a non-contact, nondestructive analytical technique used to measure surface topography. It is particularly well...
Scanning Acoustic Microscopy (SAM) is a non-destructive and non-invasive imaging technique which uses ultrasound signals to visualize the...
Scanning electron microscopy (SEM) is a surface imaging technique capable of achieving nm resolution on topographical features. Additionally,...
White light interferometry (WLI) is a nondestructive, non-contact, optical surface topography measurement which uses coherence scanning interferometry to...
Wide Area 3D Patterned Light measurements encompass a class of optical profilometry techniques used to visualize the surface...